Surgical fixation of displaced midshaft clavicle fractures: elastic intramedullary nailing versus precontoured plating
© The Author(s) 2014
Received: 7 May 2013
Accepted: 27 April 2014
Published: 25 May 2014
This prospective comparative study was done to evaluate the effectiveness of implants of different design (titanium elastic intramedullary nail versus anatomical precontoured dynamic compression plate) in treatment of displaced midshaft clavicular fractures.
Materials and methods
Sixty-six patients between 18 and 65 years of age were included in this study. They were randomized in two groups to be treated with either elastic intramedullary nail (EIN) or plate. Clinical and radiological assessments were performed at regular intervals. Outcomes and complications of both groups over 2 years of follow-up time were compared.
Length of incision, operation time, blood loss and duration of hospital stay were significantly less for the EIN group. American Shoulder and Elbow Surgeons (ASES) and Constant Shoulder scores were significantly higher (p < 0.05) in the plating group than the EIN group for the first 2 months but there was no significant difference found between the two groups regarding functional and radiological outcome at the 2-year follow-up. Significantly higher rates of refracture after implant removal (p = 0.045) in the plating group was observed. Infection and revision surgery rates were also higher in the plate group, but this difference was insignificant (p > 0.05).
EIN is a safe, minimally invasive surgical technique with a lower complication rate, faster return to daily activities, excellent cosmetic and comparable functional results, and can be used as an equally effective alternative to plate fixation in displaced midshaft clavicle fractures.
Level of evidence
KeywordsDisplaced midshaft clavicle fractures Elastic intramedullary nailing Anatomical precontoured plating
Fractures of the clavicle account for 2.6–4 % of all adult fractures, 35 % of all injuries to the shoulder girdle, and 69–82 % of these fractures occur in the middle-third [1, 2]. Displacement occurs in about 73 % of all midshaft clavicle fractures . The average age of patients sustaining a midshaft clavicular fracture is 33 years; 70 % of the patients are male . A fall or a direct blow to the shoulder, giving an axial compressive force on the clavicle, is the most common trauma mechanism of injury for any clavicular fracture [4, 5]. Displaced midshaft fractures have traditionally been treated non-operatively because of early reports suggesting that clavicular nonunions were very rare and clavicular mal-union, being of radiographic interest only, was without clinical importance [6, 7]. However, recent studies have found higher rates of delayed union, nonunion, shoulder pain, and shoulder weakness and residual pain with non-operative treatment . The indications for surgery include the need for earlier functional mobilization in the patient with an isolated injury, in addition to open fractures, floating shoulders and patients with polytrauma . For operative treatment, the available methods of fixation are fixation with Kirschner wires, pins (Rush pin, Knowles pin, Rockwood pin), plates with screws and external fixation [10–12].
This prospective comparative study was designed to compare outcomes and complications of titanium elastic intramedullary nailing and anatomically precontoured plating in displaced midshaft clavicular fractures.
Materials and methods
We conducted a prospective comparative study to compare outcomes and complications of closed displaced midshaft clavicular fractures treated with precontoured dynamic compression plates or with single titanium elastic intramedullary nails. Between July 2008 and June 2010, a total of 80 patients with closed displaced midshaft clavicular fractures were admitted in our hospital. Out of these 80 patients, 66 patients were included in this study. In this study, these patients were randomized according to inclusion and exclusion criteria into two equal groups of 33 patients, to be treated surgically with either a 3.5-mm precontoured dynamic compression plate (plate group) or with a single titanium elastic intramedullary nail fixation (EIN group).
Patients were excluded if they had fractures with marked comminution, duration of more than 4 weeks, open fractures, pre-existent morbidity of the ipsilateral arm, shoulder or hand, presence of neurovascular injury, and ipsilateral injuries.
Demographic profile of study
Precontoured plating group
Antegrade elastic nailing group
Mean age (years)
40.2 ± 11.2 (18–64)
38.9 ± 9.1 (20–62)
Mean injury time (days)
7.2 ± 3.2 (1–14)
6.9 ± 3.1 (1–13)
In both groups, arm sling support was given to all the patients for 2 weeks postoperatively. Early mobilization was started if pain permitted. Patients were encouraged to resume their normal daily activities after a 4-week postoperative period.
Operative time, length of incision, hospital stay, blood loss (calculated by the difference in the weights of the sponges pre- and postoperatively and adding volumes of suction loss), and pain visual analogue scale (0: none to 10: severe) on the first post-operative day were recorded for each patient. In follow-up visits, all patients were evaluated clinically at 1st, 2nd, 4th, 6th, 12th, 18th and 24th month to assess outcomes of fracture fixation in both groups, like fracture union time, union rate, shoulder and arm function. Shoulder function was evaluated according to the American Shoulder and Elbow Surgeons (ASES) score and Constant score, (both are 100-point scoring systems) . These scoring systems combine assessments of subjective symptoms and objective findings. In the Constant scoring system, the overall grading is excellent if the total score ranges from 90 to 100, good for 80–89, fair for 70–79, and poor if the scores are 69 or less.
Complications were recorded and compared between both groups. Non-union was defined as an unsuccessful healing of the bone after 6 months, clinically manifesting as pain at the fracture site and radiologically as a visible gap between the fracture parts. Deep infection was defined as infection requiring implant removal. Refracture was defined as a fracture of the clavicle within 3 months of implant removal without any history of retrauma.
Student’s t-test was used to analyze the difference of means for different parameters. Mean, standard deviation and standard error of mean for the variables were also calculated. The test was referenced for a two-tailed p value, and a 95 % confidence interval was constructed around a sensitivity proportion using a normal approximation method. Statistical analyses were performed using SPSS software. A value of <0.05 was considered statistically significant.
Comparison of perioperative measures and outcomes of both groups
Precontoured plating group
Antegrade elastic nailing group
Surgery time (min)
Length of incision (cm)
Pain (visual analogue scale)
Hospital stay (days)
Average blood loss (ml)
32 (100 %)
32 (96.96 %)
Union time (months)
Comparison of complications of both groups
Precontoured plating group (%)
Antegrade elastic nailing group (%)
Refracture after implant removal
Major revision surgeries
No implant failure occurred in the plate group while one implant failure was seen in the EIN group (3.03 %) (p = 0.41), which occurred within three months of the primary surgical procedure. Open reduction and plating with autogenous bone grafting in this case finally resulted in bone union. Three refractures (9.37 %) were observed in the plate group after removal of the implant without any history of fresh trauma while no such complication was seen in the EIN group (p = 0.046). All refractures occurred within 1 month after plate removal. The average age of the patients having refractures after plate removal was 37.9 years. Of these three refractures, one was treated conservatively and plating was done in two cases, leading to uneventful healing. Hypertrophic scar formation was observed in four cases in the plating group, none in the EIN group (p = 0.04); wound dehiscence was seen in three cases in the plating group and none in the EIN group (p = 0.046).
In the EIN group, elastic nails were removed in all cases. In the plate group 20 patients (total of 32 patients) underwent implant removal. In the EIN group the nail was removed at an average time of 6.2 ± 1.6 months (range 4–9 months). Plates were removed at an average time of 15.4 ± 2.2 months (range 11–20 months) (p = 0.02).
Comparison of ASES scores and Constant scores  of both groups
Precontoured plating group
Antegrade elastic nailing group
Range of motion
Total ASES score
Total Constant score
The best treatment strategy for displaced midshaft clavicle fractures remains a topic of debate. Conservative management of these fractures results in an approximately 5 % nonunion rate . While non-operative management remains the mainstay of treatment for most midshaft clavicle fractures, the indications for surgery may be expanding. Recent studies have showed a poorer outcome in cases of displaced midshaft clavicle fractures that were treated non-operatively [8, 18, 19] in comparison to surgically treated patients [16, 20, 21]. Three types of fixation are available for middle-third clavicle fractures: intramedullary devices, plates, and external fixators. Intramedullary fixation can be done by smooth or threaded K- wires, Steinman pins, Knowles pins, Hagie pins, Rush pins or cannulated screws [22–24]. Plate fixation can be done with a 3.5-mm dynamic compression plate (DCP), low-contact dynamic compression plates, reconstruction plates or locking compression plates with at least three screws (six cortices) in both the medial and lateral fragment each, and an interfragmentary lag screw whenever the fracture pattern allows it. Plating of acute clavicle fractures is advocated as the preferred fixation method by many authors [15, 25, 26]. Biomechanically, plate fixation is superior to intramedullary fixation because it better resists the bending and torsional forces that occur during elevation of the upper extremity above shoulder level . Patients treated with plate fixation can be allowed full range of motion once their soft tissues have healed. Disadvantages of plate fixation include the necessity for increased exposure and soft-tissue stripping, increased risk of damage to the supraclavicular nerve, slightly higher infection rates, and the risk of refracture after plate removal . Currently, open reduction and internal fixation with a 3.5-mm dynamic compression plate [28, 29] is the standard method; however, intramedullary fixation [16, 30] is an equally effective alternative. In this study, both methods of fixation were compared in terms of outcomes and complications.
In our study, functional shoulder scores were significantly higher for the plating group than the EIN group in the first 12 weeks, but at the 12-month follow-up visit, there was no significant difference observed between the two groups in terms of shoulder scores. In this study, in the plating group, rates of refracture (9.37 %), major revision surgery (6.25 %) and implant failure (3.03 %) were comparable to other studies. The Canadian Orthopaedic Trauma Society reported one (1.6 %) case of early mechanical failure . Böstman et al.  studied 103 patients treated with open reduction and internal fixation using plates; among those patients, 43 % had complications; 15 %, major complications; 14 % required re-operation and there was an implant failure rate of 14.6 %. Chen et al.  reported a 7.1 % implant failure rate. Liu et al.  compared titanium elastic nail and reconstruction plate fixation in displaced midshaft clavicle fractures and found no significant difference between intramedullary and plate fixation after 18 months in terms of functional outcome (DASH score p = 0.42, Constant score p = 0.17) and complications. They reported an implant failure rate of 8.5 %. In our study, the refracture rate was significantly greater in the plate group than in the EIN group. Wijdick et al.  analysed retrospectively 90 patients with displaced mid clavicle fractures treated with plate fixation or EIN. Complications were evaluated in both treatment groups and subsequently compared. Three refractures (7.0 %) were observed in the plate group after removal of the implant against none in the EIN group (p = 0.105). All refractures occurred within 2 months after removal of the implant. Poigenfurst et al.  followed 122 patients after plating of displaced clavicle fractures. There were four refractures after plate removal. The reason behind this higher refracture rate after implant removal in the plating group is that plate fixation provides a rigid fixation leading to primary bone healing: that’s why, after plate removal, the mechanical strength of the healed fracture site is reduced, explaining higher refracture rates. Along with this, screw holes may act as focal points for stress, leading to refractures. Secondary bone healing occurs in cases of fractures treated with EINs so the refracture rate after removal of the implant is lower in these cases. For plate fixation a larger incision is required, leading to a higher risk of infection and lesser cosmetic satisfaction but in our study no significant differences in infection rates between the two groups were found.
Ferran et al.  compared Rockwood pin fixation (17 cases) and low contact dynamic compression plate (LCDCP; 15 cases) in displaced midshaft clavicle fractures and found no significant difference after 12 months in functional outcome (Constant score p = 0.37). Complications occurred in 12 % of the intramedullary fixation group and in 40 % of the plate fixation group. Bohme et al.  reported the same conclusions in their study comparing plating, intramedullary fixation and conservative treatment in displaced midshaft clavicle fractures. Thyagarajan et al.  retrospectively evaluated 51 patients (three groups, each had 17 patients) with midshaft clavicle fractures. Group 1 underwent intramedullary stabilization using clavicle pins. Group 2 underwent open reduction and internal fixation using plates and group 3 underwent non-operative treatment with a sling. In group 2, two (12 %) patients had prominent hardware causing discomfort, and they underwent removal of hardware 12 months following the fixation. After implant removal results were satisfactory and there was no incidence of refracture.
In a retrospective study done by Wu et al. , comparison between plating and intramedullary nailing for the treatment of clavicular nonunion showed an 18.2 % nonunion rate with plating compared with 11.1 % for nailing, the difference being attributed to the nailʼs resistance to compressive stresses. The authors concluded that plating provides better rotational stability. Several other studies have found intramedullary fixation to be equally effective as plating, especially for the treatment of nonunion [38, 39]. Refracture after implant removal and major revision surgery just tended to prevail more often after plate fixation, while implant failure was more common in EIN groups. Major revision procedures were done in EIN groups due to implant failure, while in plating groups it was due to refracture after implant removal. Minor revision surgeries were common in EIN groups for problems like medial protrusion causing irritation or skin perforation. Major complications described in the literature for other modes of intramedullary fixation of clavicle fractures (Kirschner wire, Rush pin etc.), like injury to neurovascular structures and implant migration into the chest cavity [40, 41] were not observed in our study. No such complication has been described in the literature using TENs in clavicle fractures . Implant removal in the plating group needed another surgery done under general anesthesia, and a large-sized incision was made, while in the EIN group the nail was removed as an outdoor procedure under local anesthesia and a small incision over the tip of the nail was made. This was another advantage of intramedullary flexible nailing over plating.
The primary limitation of our study was that it was a small prospective comparative study including a small number of patients and done at a single center. Larger randomized controlled trials are needed to further evaluate outcomes and complications of precontoured plates and EIN in displaced midshaft clavicle fractures. Still, we can conclude from our study that both precontoured plating and intramedullary flexible nailing are equally effective alternatives for surgical fixation of displaced midshaft clavicular fractures. Antegrade flexible intramedullary nailing techniques have advantages like less soft tissue injury, shorter operating time and hospital stay, less blood loss, more cosmetic satisfaction and minor surgery needed to remove the implant. EIN is a safe, minimally invasive surgical technique with a lower complication rate, faster return to daily activities, excellent cosmetic and comparable functional results, which can be regard as an alternative to plate fixation of displaced midshaft clavicular fractures.
Conflict of interest
This study was carried out at a level 1 trauma centre, Mayo Institute of Medical Sciences (MIMS), India and approved by the local ethical committee and institutional review board of MIMS. All patients gave their informed consent. This was performed in accordance with the ethical standards of the 1964 Declaration of Helsinki as revised in 2000.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution License which permits any use, distribution, and reproduction in any medium, provided the original author(s) and the source are credited.
- Robinson CM (1998) Fractures of the clavicle in the adult. Epidemiology and classification. J Bone Joint Surg Br 80:476–484PubMedView ArticleGoogle Scholar
- Khan LA, Bradnock TJ, Scott C, Robinson CM (2009) Fractures of the clavicle. J Bone Joint Surg (Am) 91:447–460View ArticleGoogle Scholar
- Pearson AM, Tosteson ANA, Koval KJ, McKee MD, Cantu RV, Bell JE, Vicente M (2010) Is surgery for displaced, midshaft clavicle fractures in adults cost-effective? Results based on a multicenter randomized, controlled trial. J Orthop Trauma 24:426–433PubMedView ArticleGoogle Scholar
- Nowak J, Mallmin H, Larsson S (2000) The aetiology and epidemiology of clavicular fractures. A prospective study during a two-year period in Uppsala, Sweden. Injury 31:353–358PubMedView ArticleGoogle Scholar
- Stanley D, Trowbridge EA, Norris SH (1988) The mechanism of clavicular fracture. A clinical and biomechanical analysis. J Bone Joint Surg Br 70:461–464PubMedGoogle Scholar
- Neer CS (1960) Nonunion of the clavicle. J Am Med Assoc 172:1006–1101PubMedView ArticleGoogle Scholar
- Canadian Orthopaedic Trauma Society (2007) Nonoperative treatment compared with plate fixation of displaced midshaft clavicular fractures. A multicenter, randomized clinical trial. J Bone Joint Surg Am 89:1–10Google Scholar
- Hill JM, McGuire MH, Crosby LA (1997) Closed treatment of displaced middle-third fractures of the clavicle gives poor results. J Bone Joint Surg Br 79:537–539PubMedView ArticleGoogle Scholar
- Denard PJ, Koval KJ, Cantu RV, Weinstein JN (2005) Management of midshaft clavicle fractures in adults. Am J Orthop 34:527–536PubMedGoogle Scholar
- Lee YS, Lin CC, Huang CR, Chen CN, Liao WY (2007) Operative treatment of midclavicular fractures in 62 elderly patients: Knowles pin versus plate. Orthopedics 30(11):959–964PubMedGoogle Scholar
- Mudd CD, Quigley KJ, Gross LB (2011) Excessive complications of open intramedullary nailing of midshaft clavicle fractures with the Rockwood clavicle pin. Clin Orthop Relat Res 469(12):3364–3370PubMed CentralPubMedView ArticleGoogle Scholar
- Mueller M, Rangger C, Striepens N, Burger C (2008) Minimally invasive intramedullary nailing of midshaft clavicular fractures using titanium elastic nails. J Trauma 64(6):1528–1534PubMedView ArticleGoogle Scholar
- Assobhi JE (2011) Reconstruction plate versus minimal invasive retrograde titanium elastic nail fixation for displaced midclavicular fractures. J Orthop Traumatol 12(4):185–192PubMed CentralPubMedView ArticleGoogle Scholar
- Wijdicks FJG, Houwert M, Dijkgraaf M, de Lange D, Oosterhuis K, Clevers G, Millet PJ, Verleisdonk EJ (2012) Complications after plate fixation and elastic stable intramedullary nailing of dislocated midshaft clavicle fractures: a retrospective comparison. Int Orthop 36(10):2139–2145PubMed CentralPubMedView ArticleGoogle Scholar
- Poigenfürst J, Rappold G, Fisher W (1992) Plating of fresh clavicular fractures: results of 122 operations. Injury 23(4):237–241PubMedView ArticleGoogle Scholar
- Jubel A, Andermahr J, Schiffer G, Tsironis K, Rehm KE (2003) Elastic stable intramedullary nailing of midclavicular fractures with a titanium nail. Clin Orthop Relat Res 408:279–285PubMedView ArticleGoogle Scholar
- Constant CR, Murley AG (1987) A clinical method of functional assessment of the shoulder. Clin Orthop Relat Res 214:160–164PubMedGoogle Scholar
- Robinson CM, Court-Brown CM, McQueen MM, Wakefield AE (2004) Estimating the risk of nonunion following nonoperative treatment of a clavicular fracture. J Bone Joint Surg 86-A:1359–1365PubMedGoogle Scholar
- Wild LM, Potter J (2006) Deficits following nonoperative treatment of displaced midshaft clavicular fractures. J Bone Joint Surg 88-A:35–40Google Scholar
- Ali KM, Lucas HK (1978) Plating of fractures of the middle third of the clavicle. Injury 9(4):263–267Google Scholar
- Jeray K (2007) Acute midshaft clavicular fracture. J Am Acad Orthop Surg 15:239–248PubMedGoogle Scholar
- Kyle RF, Schmidt AH (1995) Open reduction and internal fixation of fractures and nonunions. In: Thompson RC (ed) The shoulder. Raven Press, New York, p 183Google Scholar
- Neviaser RJ, Neviaser JS, Neviaser TJ (1975) A simple technique for internal fixation of the clavicle: a long-term evaluation. Clin Orthop 109:103PubMedView ArticleGoogle Scholar
- Ngarmukos C, Parkpian V, Patradul A (1998) Fixation of fractures of the midshaft of the clavicle with Kirschner wires: results in 108 patients. J Bone Joint Surg 80B:106View ArticleGoogle Scholar
- Mullaji AB, Jupiter JB (1994) Low-contact dynamic compression plating of the clavicle. Injury 25:41PubMedView ArticleGoogle Scholar
- Ring D, Jupiter JB, Miller ME, Ada JR (1998) Injuries to the shoulder girdle: part II. Fractures of the clavicle. In: Browner BD, Jupiter JB, Levine AM, Trafton PG (eds) Skeletal trauma, vol 2. WB Saunders, Philadelphia, p 1670Google Scholar
- Golish SR, Oliviero JA, Francke EI, Miller MD (2008) A biomechanical study of plate versus intramedullary devices for midshaft clavicle fixation. J Orthop Surg Res 16(3):28View ArticleGoogle Scholar
- Kloen P, Sorkin AT, Rubel IF, Helfet DL (2002) Anteroinferior plating of midshaft clavicular nonunions. J Orthop Trauma 16:425–430PubMedView ArticleGoogle Scholar
- Schwarz N, Hocker K (1992) Osteosynthesis of irreducible fractures of the clavicle with 2.7-mm ASIF plates. J Trauma 33:179–183PubMedView ArticleGoogle Scholar
- Rehm KE, Andermahr J, Jubel A (2005) Intramedullary nailing of midclavicular fractures with an elastic titanium nail. Eur J Traum Emerg Surg 31(4):409–416Google Scholar
- Böstman O, Manninen M, Pihlajamäki H (1997) Complications of plate fixation in fresh displaced midclavicular fractures. J Trauma 43:778–783PubMedView ArticleGoogle Scholar
- Chen YF, Wei HF, Zhang C, Zeng BF, Zhang CQ, Xue JF et al (2012) Retrospective comparison of titanium elastic nail (TEN) and reconstruction plate repair of displaced midshaft clavicular fractures. J Shoulder Elbow Surg 21:495–501PubMedView ArticleGoogle Scholar
- Liu HH, Chang CH, Chia WT, Chen CH, Tarng YW, Wong CY (2010) Comparison of plates versus intramedullary nails for fixation of displaced midshaft clavicular fractures. J Trauma 69(6):E82–E87PubMedView ArticleGoogle Scholar
- Ferran NA, Hodgson P, Vannet N, Williams R, Evans RO (2010) Locked intramedullary fixation vs plating for displaced and shortened mid-shaft clavicle fractures: a randomized clinical trial. J Shoulder Elbow Surg 19:783–789PubMedView ArticleGoogle Scholar
- Böhme J, Bonk A, Bacher GO, Wilharm A, Hoffmann R, Josten C (2010) Current treatment concepts for mid-shaft fractures of the clavicle—results of a prospective multicentre study. Z Orthop Unfall. 149(1):68–76PubMedGoogle Scholar
- Thyagarajan DS, Day M, Dent C, Williams R, Evans R (2009) Treatment of mid-shaft clavicle fractures: a comparative study. Int J Shoulder Surg 3(2):23–27PubMed CentralView ArticleGoogle Scholar
- Wu CC, Shih CH, Chen WJ, Tai CL (1998) Treatment of clavicular aseptic nonunion: comparison of plating and intramedullary nailing techniques. J Trauma 45:512–516PubMedView ArticleGoogle Scholar
- Capicotto PN, Heiple KG, Wilbur JH (1994) Midshaft clavicle nonunions treated with intramedullary Steinman pin fixation and onlay bone graft. J Orthop Trauma 8:88PubMedView ArticleGoogle Scholar
- Enneking TJ, Hartlief MT, Fontijne WP (1999) Rush pin fixation for midshaft clavicular nonunions: good results in 13/14 cases. Acta Orthop Scand 70:514–516PubMedView ArticleGoogle Scholar
- Schwarz N, Leixnering M (1984) Failures of clavicular intramedullary wire fixation and their causes. Aktuelle Traumatol 14:159–163PubMedGoogle Scholar
- Liu HP, Chang CH, Lin PJ, Chu JJ, Hsieh HC, Chang JP et al (1993) Pulmonary artery perforation after Kirschner wire migration: case report and review of the literature. J Trauma 34:154–156PubMedView ArticleGoogle Scholar