# Systematized water content calculation in cartilage using T1-mapping MR estimations: design and validation of a mathematical model

- J. M. Shiguetomi-Medina
^{1}Email author, - J. L. Ramirez-GL
^{2}, - H. Stødkilde-Jørgensen
^{3}and - B. Møller-Madsen
^{4}

https://doi.org/10.1007/s10195-016-0433-8

© The Author(s) 2016

**Received: **7 June 2016

**Accepted: **8 October 2016

**Published: **22 October 2016

## Abstract

### Background

Up to 80 % of cartilage is water; the rest is collagen fibers and proteoglycans. Magnetic resonance (MR) T1-weighted measurements can be employed to calculate the water content of a tissue using T1 mapping. In this study, a method that translates T1 values into water content data was tested statistically.

### Materials and methods

To develop a predictive equation, T1 values were obtained for tissue-mimicking gelatin samples. 1.5 T MRI was performed using inverse angle phase and an inverse sequence at 37 (±0.5) °C. Regions of interest were manually delineated and the mean T1 value was estimated in arbitrary units. Data were collected and modeled using linear regression. To validate the method, articular cartilage from six healthy pigs was used. The experiment was conducted in accordance with the Danish Animal Experiment Committee. Double measurements were performed for each animal. Ex vivo, all water in the tissue was extracted by lyophilization, thus allowing the volume of water to be measured. This was then compared with the predicted water content via Lin’s concordance correlation coefficient at the 95 % confidence level.

### Results

The mathematical model was highly significant when compared to a null model (*p* < 0.0001). 97.3 % of the variation in water content can be explained by absolute T1 values. Percentage water content could be predicted as 0.476 + (T1 value) × 0.000193 × 100 %. We found that there was 98 % concordance between the actual and predicted water contents.

### Conclusion

The results of this study demonstrate that MR data can be used to predict percentage water contents of cartilage samples.

### Level of evidence

3 (case-control study).

### Keywords

Water content MR scanning Mathematical model## Introduction

A mathematical model can be defined as a mathematical representation of the behavior of a real device or object [1]. In science, it is essential to model phenomena because these models can be used to explain behavior, thus allowing predictions to be made [2]. Cartilage is an intricate balance of water, chondrocytes, and a rich extracellular matrix of collagen fibers and proteoglycan molecules. Up to 80 % (by weight) of this tissue is water [3]. As cartilage is mainly composed of water, the water content of the cartilage is an indicator of cartilage health. Using magnetic resonance (MR), it is possible to accurately describe morphology and tissue integrity. Some authors consider this technique the new gold standard for cartilage image analysis [3, 4]. It has been reported that disease affects the water content of cartilage even before clinical or radiological signs appear [5]. Therefore, it is important to quantify the water content in this tissue as the water content can then be used to diagnose disease and estimate its prognosis [6]. T1-weighted magnetic resonance measurements provide enough data to allow the water content to be calculated. This method has been used in several scientific fields; see for instance [7–10], although these reports only provide information on the intensity of the T1 signal. In the present study, a similar method was developed to quantify the T1 signal in cartilage [11]. To our knowledge, this is the first statistically proven method of translating T1-mapping values into clinically useful data (water content as a percentage or in milliliters) to be published. The aims of this study were therefore to develop an equation that could be used to estimate the percentage water content of cartilage tissue based on T1-weighted MR images, and to assess the performance of this equation in vivo.

## Materials and methods

### Development of the water content equation

### Concordance between the predicted and real water contents

To assess the performance of the predictive equation for a sample of healthy cartilage, the tibial proximal growth plates of six healthy, skeletally immature, 30-kg Danish Landrace pigs were used. The experiment was conducted in accordance with the Danish Animal Research Guidelines, and was accepted and authorized by the Danish Animal Experiment Committee (study J. No. 2008/561-329). The animals received an intramuscular injection of ketaminol (5 mg/kg* S*-ketamine, Pfizer, Berlin, Germany) and midazolam (0.5 mg/kg, Hypnomidate, Janssen–Cilag, Beerse, Belgium). The animals were intubated and connected to a ventilator (frequency rate 16, volume 400 ml, 30 % oxygen). Anesthesia was maintained with intravenous infusion of propofol (5 mg/kg/h, Fresenius Kabi AB, Uppsala, Sweden) and fentanyl (0.025 mg/kg/h, Haldid, Janssen–Cilag). Measurements were performed in both hind legs of each animal on the same scan using the MR protocols discussed above. The area of interest was manually delineated on the growth plate and the mean T1 values were obtained. The T1 values were inputted into the water content equation, and the estimated percentage water content was calculated. Right after euthanasia, the tibiae were harvested and the proximal growth plates were isolated. All water in the tissue was extracted from each growth plate by lyophilization (freeze-drying), allowing us to measure the actual volume of water in the tissue. The difference between the wet and dry weights was used to estimate the percentage water content. The actual water content was compared to the predicted water content using Lin’s concordance correlation coefficient, which has been shown to be superior to other correlation coefficients in paired data analysis [12, 13]. The statistical package R v.3.0.2 was used [14] and the 95 % confidence level was derived.

## Results

### Water content equation

*p*< 0.0001,

*r*

^{2}= 0.973). It was adjusted by 150 bootstrap repetitions to an

*r*

^{2}-corrected index of 0.9715 (Fig. 2). In the equation, when the mean absolute T1 value increases by one unit, the water content increases by 0.019 % (95 % CI 0.018–0.020,

*p*< 0.0001). The final equation for the predicted percentage water content was: % water content = (0.476 + T1 signal intensity × 0.000193) × 100 %.

### Water content concordance

## Discussion

The evaluation of cartilage using magnetic resonance data provides information on the integrity and physiological condition of tissue; specifically, its water content [11]. Our results allow the T1 value of any cartilage sample to be converted into a percentage water content through the use of a predictive equation. We believe that such an approach is useful in the clinical arena since it provides data that can be easily handled and translated into clinically relevant information. T1 values provide information on the water content because the protons in the cartilage interact with the water molecules that surround them [15]. Since this lattice consists mainly of water [3], using inverse sequence echo gradients allows the free water component to be differentiated from the protein-bound water component [15]. The content of free water is an indicator of the health of the tissue [5]. Because absolute T1 values are considered in this approach, the predictive equation could be used for values obtained under other magnetic fields (we performed our experiments under a 1.5 T magnetic field). However, a specific experiment to prove this should be conducted. The predictive equation developed in this study was shown in the in vivo assays to predict the true water content with 98 % accuracy. The clinical significance of the quantification of cartilage water content is the early identification of patients who present degeneration of the proteoglycan matrix as well as tissue hydration even before clinical symptoms occur. This could allow the development of predictive parameters for cartilage health and the establishment of early therapy in the future. The main limitation of this study is that a small sample of real tissue was used for the correlation assay. Although our results appear to be robust, they cannot be used to make clinical predictions. Clinical implications of this technique may be the assessment of subclinical damage in joint cartilage, the assessment of water content in intervertebral discs, and the development of prognostic indicators for osteoarthritis and cartilage injuries. Gelatine samples resemble not only the cartilage lattice but also other tissues with high water concentrations, meaning that this equation could be applied to those tissues too, thus allowing the identification of other diseases, although further studies must be carried out to verify this. They may be multiple applications of this approach beyond the field of musculoskeletal diseases, and tests assessing whether the predictive equation is useful in tissues with different lattices should be performed. Potential applications of this technique could be the quantification of cerebral edema in patients with traumatic brain injury, the quantification of water volume in patients with edematous states, and the estimation of blood volume in a tumor. Also, most newly diagnosed patients are not in the early stages of disease. This equation could be used in an objective follow-up to treatment or disease progression. Many orthopedic diseases involve cartilage; this equation could help with diagnosis and treatment. The utilization of the equation must be approved by the treating orthopedic surgeon. We are proposing a new tool.

In conclusion, T1 values of cartilage obtained through inverse angle phase and reverse sequence retrieval methods provide information on the water content of tissue. These values can be translated into percentage water contents using a predictive formula with an in vivo accuracy of 98 %.

## Declarations

### Compliance with ethical standards

### Conflict of interest

All of the authors declare that they have no conflict of interest.

### Ethical standards

All applicable international, national, and/or institutional guidelines for the care and use of animals were followed.

**Open Access**This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.

## Authors’ Affiliations

## References

- Dym CL (2004) Principles of mathematical modeling, 2nd edn. Academic, Dublin, p 303. http://www.amazon.com/Principles-Mathematical-Modeling-Second-Edition/dp/0122265513. Accessed 10 Dec 2015
- Carson E, Cobelli C (2013) Modelling methodology for physiology and medicine, 2nd edn. Elsevier, Amsterdam, p 588. http://www.amazon.com/Modeling-Methodology-Physiology-Medicine-Elsevier/dp/0124115578/ref=sr_1_2?ie=UTF8&qid=1412004935&sr=8-2&keywords=Modelling+Methodology+for+Physiology+and+Medicine. Accessed 10 Dec 2015
- Jazrawi LM, Alaia MJ, Chang G, Fitzgerald EF, Recht MP (2011) Advances in magnetic resonance imaging of articular cartilage. J Am Acad Orthop Surg 19(7):420–429View ArticlePubMedGoogle Scholar
- Shiguetomi-Medina JM, Rahbek O, Ringgaard S, Kristiansen MS, Stødkilde-Jørgensen H, Møller-Madsen B (2014) Accuracy of MR in growth plate measurement. Skelet Radiol 43(9):1263–1269View ArticleGoogle Scholar
- Xie L, Lin ASP, Kundu K, Levenston ME, Murthy N, Guldberg RE (2012) Quantitative imaging of cartilage and bone morphology, reactive oxygen species, and vascularization in a rodent model of osteoarthritis. Arthritis Rheum 64(6):1899–1908View ArticlePubMedPubMed CentralGoogle Scholar
- Cutillo AG, Morris AH, Ailion DC, Durney CH, Ganesan K (1992) Determination of lung water content and distribution by nuclear magnetic resonance. New aspects on respiratory failure. Springer, Berlin, pp 138–46. doi:https://doi.org/10.1007/978-3-642-74943-8
- Fatouros PP, Marmarou A, Kraft KA, Inao S, Schwarz FP (1991) In vivo brain water determination by T1 measurements: effect of total water content, hydration fraction, and field strength. Magn Reson Med 17(2):402–413Google Scholar
- Müller-Petke M, Dlugosch R, Yaramanci U (2011) Evaluation of surface nuclear magnetic resonance-estimated subsurface water content. New J Phys 6(13):095002View ArticleGoogle Scholar
- Richardson C, Keller EE (1966) The brine content of sea ice measured with a nuclear magnetic resonance spectrometer. J Glaciol 6(43):89–100Google Scholar
- Bae WC, Chen PC, Chung CB, Masuda K, D’Lima D, Du J (2012) Quantitative ultrashort echo time (UTE) MRI of human cortical bone: correlation with porosity and biomechanical properties. J Bone Miner Res 27(4):848–857Google Scholar
- Shiguetomi-Medina JM, Gottliebsen M, Kristiansen MS, Ringgaard S, Stødkilde-Jørgensen H, Rahbek O et al (2013) Water-content calculation in growth plate and cartilage using MR T1-mapping design and validation of a new method in a porcine model. Skelet Radiol 42(10):1413–1419View ArticleGoogle Scholar
- Lin LI (1989) A concordance correlation coefficient to evaluate reproducibility. Biometrics 45(1):255–268View ArticlePubMedGoogle Scholar
- Lin LI-K (1992) Assay validation using the concordance correlation coefficient. Biometrics 48(2):599View ArticleGoogle Scholar
- R Core Team (2014) R: a language and environment for statistical computing. R Foundation for Statistical Computing, Vienna. http://www.R-project.org/
- Currie S, Hoggard N, Craven IJ, Hadjivassiliou M, Wilkinson ID (2013) Understanding MRI: basic MR physics for physicians. Postgrad Med J 89(1050):209–223View ArticlePubMedGoogle Scholar