Dissociation of modular total hip arthroplasty at the neck–stem interface without dislocation
© The Author(s) 2011
Received: 26 February 2011
Accepted: 15 November 2011
Published: 8 December 2011
Modular femoral and acetabular components are now widely used, but only a few complications related to the modularity itself have been reported. We describe a case of dissociation of the modular total hip arthroplasty (THA) at the femoral neck–stem interface during walking. The possible causes of this dissociation are discussed. Successful treatment was provided with surgical revision and replacement of the modular neck components. Surgeons who use modular components in hip arthroplasties should be aware of possible early complications in which the modularity of the prostheses is the major factor of failure.
KeywordsTotal hip arthroplasty Femoral neck–stem dissociation Surgical revision
Modular femoral components have the advantages of reducing the need to stock many stem and head sizes and of allowing the final choice of neck length and head size to be made after stem implantation. Neck orientation can also be changed after implantation, which as is well known can be the cause of early dislocation. The incidence of postoperative dislocation of modular total hip arthroplasty (THA) varies from 0.5% to 4% . Dissociation at the neck–stem interface is rare. To the best of our knowledge, only three case reports have been published [2–4], but they pertained to dissociation at the neck–head interface. We report a case of dissociation at the neck–stem interface without hip dislocation that occurred during walking, and we discuss the causes of dissociation as well as strategies to avoid and treat this complication.
The wound healed smoothly after operation, and Harris Hip Score functional score was 90.
The use of modular components greatly increases flexibility during THA but also introduces the risk of failures at the interfaces and possible intraoperative errors in matching. Dislocation is a potential problem after THA [1, 5, 6], and dissociation of modular components after dislocation is unique to modular systems. Dissociation can occur during closed reduction of dislocation at two different interface levels: the fixed acetabular shell–polyethylene liner interface [2, 7–13], and the femoral head–neck interface [2–4]. In the case reported here, dissociation occurred at the femoral neck–stem interface, with no previous traumatic incidence. To the best of our knowledge, no such case has been reported previously concerning this type of prostheses. The manner in which this incident occurred reveals inadequate modular component fixation or a repetitive force that provoked micromovement of the modular interface that finally led to component dissociation. Potential causes of dissociation during normal walking are:
1. Inadequate orientation of femoral neck resulting in stress forces at the stem–neck interface; in this case, orientation of the femoral and acetabular components cannot be reliably evaluated due to the absence of a computed tomography (CT) scan of the indexed hip.
2. Excessive telescopic movements, which finally lead to dissociation by creating negative pressure in the acetabular area. Computer-assisted measurement of distal stem migration showed a subsidence of 3.6 mm at 3 months, which is considered excessive for this short postoperative period, though it is expected for this type of revision stem and transfemoral approach  (Fig. 4). Such an early stem subsidence and subsequent leg shortening can result in loss of intraoperative soft tissue tension and, eventually, in hip-joint instability.
3. Impingement of the femoral neck at the acetabular shell or at osteophytes in the area, causing mechanical stresses at the finally dissociated interface. As mentioned above, component to component impingement cannot be confirmed in our case. However, we consider bony impingement to be more important for this patient. Arc length between the tip of the greater trochanter and the ilium (GT arc) has been shown to correlate with free hip flexion and abduction before impingement . In this case, minimal arc length and the high position of the tip of the greater trochanter in relation to the head center, predicts early bony impingement (greater trochanter to ilium) (Fig. 5). In a computer model, it has been shown that once bony impingement becomes the restricting factor, further changes in implant design and orientation may not improve range of motion (ROM) . Furthermore, in a cadaver study of hip dislocation, osseous impingement was likely to occur between the greater trochanter and the iliac wing before component impingement . Similarly, bony impingement preceded component impingement in about 44% of all conditions tested in a three-dimensional computer model with varying orientations of the femoral and acetabular components .
4. Ectopic bone formation causing abnormal movement of the joint. Heterotopic ossification can cause hip-joint instability when the periarticular bone mass limits femoral excursion or contributes to impingement . However, to our knowledge only in two cases was hip dislocation directly attributed to heterotopic ossification .
In this case, a jumbo cup was used due to extensive bone loss to ensure stable primary fixation. Three cancellous screws were also placed for the same reason. Regarding the femur, the main goal was successful diaphyseal fixation of the stem, so a long, fully porous-coated trapezoid-shaped stem was used. For the modular neck, a straight 0° long neck was selected, allowing fine positioning of the stem in relation to the cup. Although unnecessary , three medium hammer blows were applied to fix the neck–stem coupling. Intraoperatively, during the second revision, a large amount of ectopic bone was found in the lesser trochanter area (Fig. 4b), which is a possible cause of stem impingement and, in particular, the neck stem interface, which may lead to dissociation due to repetitive stresses and micromovement in the area. The ectopic bone was removed (Fig. 3), and intraoperative mobilization revealed free movement of the hip joint in all possible directions.
Modular components give the surgeon an intraoperative advantage but also increase the potential for component mismatch and mechanical failure. Dissociation is a rare but possible cause of failure. To prevent this complication, the femoral neck component should be impacted firmly onto the tapered stem base during the operation. Finally, free movement of the joint is essential to prevent abnormal stresses at the interfaces of the modular components.
Conflict of interest
This article is distributed under the terms of the Creative Commons Attribution License which permits any use, distribution and reproduction in any medium, provided the original author(s) and source are credited.
- Carlsson AS, Gentz CF (1977) Postoperative dislocation in the Charnley and Brunswik total hip arthroplasty. Clin Orthop 125:177–182PubMedGoogle Scholar
- Star MJ, Colwell CW Jr, Donaldson WF III, Walker RH (1992) Dissociation of modular hip arthroplasty components after dislocation: a report of three cases at differing dissociation levels. Clin Orthop 278:111–115PubMedGoogle Scholar
- Chu CM, Wang SJ, Lin LC (2001) Dissociation of modular total hip arthroplasty at the femoral head–neck interface after loosening of the acetabular shell following hip dislocation. J Arthroplast 16(6):806–809View ArticleGoogle Scholar
- Woolson ST, Pottorff GT (1990) Disassembly of a modular femoral prosthesis after dislocation of the femoral component: a case report. J Bone Joint Surg Am 72(4):624–625PubMedGoogle Scholar
- Gill TJ, Sledge JB, Müller ME (1998) The Bürch-Schneider anti-protrusio cage in revision total hip arthroplasty: indications, principles and long-term results. J Bone Joint Surg Br 80(6):946–953PubMedView ArticleGoogle Scholar
- Lewinnek GE, Lewis JL, Tarr R, Compere CL, Zimmerman JR (1978) Dislocations after total hip-replacement arthroplasties. J Bone Joint Surg Am 60(2):217–220PubMedGoogle Scholar
- Cameron HU (1993) Dissociation of a polyethylene liner from an acetabular cup. Orthop Rev 22(10):1160–1161PubMedGoogle Scholar
- Wilson AJ, Monsees B, Blair VP III (1988) Acetabular cup dislocation: a new complication of total joint arthroplasty. AJR 151(1):133–134PubMedView ArticleGoogle Scholar
- Kitziger KJ, DeLee JC, Evans JA (1990) Disassembly of a modular acetabular component of a total hip-replacement arthroplasty. A case report. J Bone Joint Surg Am 72(4):621–623PubMedGoogle Scholar
- Ries MD, Collis DK, Lynch F (1992) Separation of the polyethylene liner from acetabular cup metal backing. A report of three cases. Clin Orthop Relat Res 282:164–169PubMedGoogle Scholar
- Bueche MJ, Herzenberg JE, Stubbs BT (1989) Dissociation of a metal-backed polyethylene acetabular component. A case report. J Arthroplast 4(1):39–41View ArticleGoogle Scholar
- Ferenz CC (1988) Polyethylene insert dislocation in a screw-in acetabular cup. A case report. J Arthroplast 3(3):201–204View ArticleGoogle Scholar
- O’Brien RF, Chess D (1992) Late disassembly of a modular acetabular component. A case report. J Arthroplast 7 Suppl:453–455View ArticleGoogle Scholar
- Artiaco S, Boggio F, Titolo P, Zoccola K, Bianchi P, Bellomo F (2011) Clinical experience in femoral revision with the modular Profemur R stem. Hip Int 21(1):39–42PubMedView ArticleGoogle Scholar
- Bunn A, Colwell CW Jr, D’Lima DD (2011) Bony impingement limits design-related increases in hip range of motion. Clin Orthop Relat Res. (in press) [Epub ahead of print]Google Scholar
- Bartz RL, Nobel PC, Kadakia NR, Tullos HS (2000) The effect of femoral component head size on posterior dislocation of the artificial hip joint. J Bone Joint Surg Am 82(9):1300–1307PubMedGoogle Scholar
- Kessler O, Patil S, Wirth S, Mayr E, Colwell CW Jr, D’Lima DD (2008) Bony impingement affects range of motion after total hip arthroplasty: a subject-specific approach. J Orthop Res 26(4):443–452PubMedView ArticleGoogle Scholar
- Iorio R, Healy WL (2002) Heterotopic ossification after hip and knee arthroplasty: risk factors, prevention, and treatment. J Am Acad Orthop Surg 10(6):409–416PubMedGoogle Scholar
- Cobb TK, Berry DJ, Wallrichs SL, Ilstrup DM, Morrey BF (1999) Functional outcome of excision of heterotopic ossification after total hip arthroplasty. Clin Orthop Relat Res 361:131–139PubMedView ArticleGoogle Scholar
- Grupp TM, Weik T, Bloemer W, Knaebel HP (2010) Modular titanium alloy neck adapter failures in hip replacement—failure mode analysis and influence of implant material. BMC Musculoskelet Disord 11:3PubMed CentralPubMedView ArticleGoogle Scholar
- Toni A, Sudanese A, Paderni S, Guerra E, Bianchi G, Antonietti B, Giunti A (2001) Cementless hip arthroplasty with a modular neck. Chir Organi Mov 86(2):73–85PubMedGoogle Scholar
- Pallini F, Cristofolini L, Traina F, Toni A (2007) Modular hip stems: determination of disassembly force of a neck-stem coupling. Artif Organs 31(2):166–170PubMedView ArticleGoogle Scholar