The migration pattern of the Charnley femoral stem: a five-year follow-up RSA study in a well-functioning patient group
© The Author(s) 2012
Received: 1 July 2011
Accepted: 10 March 2012
Published: 11 May 2012
Implant stability is considered vital to long-time implant survival in total hip arthroplasty (THA), since loose implants are reported to be a major cause of hip revision. There is an association between early implant micromotion and increased risk of revision. More implant-specific data are needed to establish acceptable levels of early implant movement.
Materials and methods
Thirty-five patients (36 hips) undergoing Charnley THA were followed with repeated clinical, radiographic, and radiostereometric analysis (RSA) over 5 years. Twenty-three patients attended 5 years postoperatively.
The patient group was well functioning based on the radiological and clinical evaluations. The stems constantly moved up to 5 years postoperatively, with subsidence, retroversion, and varus tilt, based on the RSA.
Continuous movement of the Charnley stem was observed up to 5 years postoperatively in a well-functioning patient group. The migration data presented herein could be useful when defining acceptable migration limits for certain types of cemented femoral stems.
KeywordsRSA Charnley THR Five-year follow-up Migration pattern
Implant stability influences the revision rate observed in total hip arthroplasty (THA). According to the Swedish, Australian and Norwegian hip registers, aseptic loosening is the most common cause of revision [1–3]. Revision surgery is expensive to both patients and the community. Major efforts have been made to reduce these numbers from the earliest days of THA.
Radiostereometric analysis (RSA) has become a gold standard when measuring implant migrations with respect to the cement mantle as well as the surrounding bone . RSA can describe early implant migration, and is therefore well suited and recommended when introducing new implants into clinical use and minimizing the number of enrolled patients .
The main rationale for monitoring early implant migration is based on results from clinical studies reporting an association between early subsidence and increased risk of early or mid-term hip implant revision. Early migration , continuous migration , and subsidence in combination with medial and posterior migration  are of concern. However, when defining acceptable limits for migration, factors such as measurement procedures, implant design, and fixation principles should be taken into account.
The level of early implant stability is mainly determined by the fixation method used. In cement fixation of femoral stems, two different principles apply [8, 9]. In a composite construction, such as the Charnley stem, there should be perfect bonding between stem and cement. In contrast, there is the construction in which the stem is not intended to be bonded to the cement, but rather to act as a “loaded taper” that is constrained by its surrounding cement mantle, such as the Exeter stem.
With uncemented fixation, primary stability is most important, as it facilitates osseointegration and thereby secondary stability.
Both uncemented stems and cemented loaded taper stems are meant to be somewhat “loose” shortly after surgery. A relatively high amount of migration is consequently expected when the patients start weight-bearing. In contrast, the composite construction should be mechanically stable after setting the cement. Only a small amount of migration is then to be expected during early postoperative rehabilitation.
When levels of early migration are expected to vary due to differences in fixation, implant design, etc., it could be wise to disregard early migration data so as to make study comparisons more comprehensive. Early postoperative RSA measurements typically serve as a baseline for measures of implant movement. To overcome expected differences in early implant movements, 2 or 3 years of data have been suggested for use as the baseline when comparing levels of migration .
The rationale of predicting medium- to long-term prosthetic loosening based on short-term migration data is based on publications that focus on different implants and different measurement techniques [6, 7, 11, 12].
Implant-specific migration data could be useful to improve the prediction of late implant loosening based on short-term migration data. Background data from well-functional, nonrevised total hip replacements are then needed to establish acceptable limits of femoral stem migration. In the present RSA study, we describe the migration pattern of the Charnley hip stem in a group of well-functioning patients at 5-year follow-up.
Materials and methods
The study was approved by the Norwegian Technical and Scientific University Central Region Ethics Committee (reference 094-02), and conducted in accordance with the Declaration of Helsinki.
Originally, this study was designed as a 2-year prospective randomized cohort RSA study to investigate the stability of the Charnley flange 40 femoral stem prosthesis fixated with either Palacos R or SmartSet HV cement. No significant difference in stem fixation was found at the 2-year follow-up . According to the clinical evaluation, all patients were performing very well clinically and the study was extended to include a 5-year follow-up. The extension was approved by the Ethics Committee. The main purpose of the study was then to describe the migration pattern of the Charnley stem, regardless of the type of cement employed.
The patients were recruited from Trondheim University Hospital and operated on by one experienced surgeon (OSH). Thirty-five patients, representing 36 hips, were asked, and all agreed to participate. Informed patient consent was signed on the day of surgery.
Patients with an existing condition, such as malignancy, pregnancy, severe osteoporosis, corticosteroid treatment, disabling musculoskeletal problems (other than in the hips), as well as patients who had already participated in a clinical study with an investigational product within the last 6 months were not invited to participate. No exclusions were made based on these conditions.
The 23 patients were operated between October 2002 and October 2003. A bilateral procedure where both hips received the Palacos cement was performed in one patient. Mean (range) age and mean (range) weight at surgery were 69 (60–76) years and 75 (60–98) kg, respectively. At five-year follow-up 22 patients were diagnosed with osteoarthritis and one patient with congenital hip dysplasia.
The cement randomization was performed preoperatively by a computer program (SAS version 8). An equal number of hips was allocated to each cement group. No stratification based on age or gender was made. The cement type to be used was revealed peroperatively. Differences in color and handling made the surgeon aware of the cement type allocated, while the patients were kept unaware.
A posterolateral surgical approach was used, with a lateral incision. The femoral canal was opened through the piriformis fossa and prepared with a 12 mm central reamer, followed by a 1 mm all-direction oversized broach. Intrafemoral pulsed lavage followed by a temporary 1 % adrenalin-soaked sponge were applied to prepare the cavity prior to cementing.
We performed the RSA follow-up according to the Selvik method . Nine tantalum beads (0.8 mm in diameter) were implanted to serve as references: ideally five in the greater trochanter and four in the lesser trochanter region. The initial RSA examination was performed within 1 week postoperatively, after weight bearing (Table 1).
The RSA measurements were performed using UmRSA software (version 6.0, RSA Biomedical Innovation, Umea, Sweden). The tantalum balls had to be sufficiently scattered within each segment to ensure acceptable quality measurement. The condition number (CN) is a measure of the configuration of the markers in the segment. The mean error of rigid body fitting (ME) is a measure of marker stability. The upper limit on the ME was set to 0.25 mm, and the upper limit for the CN was set to 130, according to the guidelines in theInstructional RSA User Course Manuala(RSA Biomedical). The ME and CN were met in all analyses except for 1. This patient was excluded. The results for the present study are presented in Table 2.
Several RSA studies conclude that the migration of a cemented femoral stem mainly occurs at the implant–cement interface, not between the cement mantle and surrounding bone [15–17]. Tantalum markers that are mixed into the cement are difficult to scatter and also difficult to visualize on radiographs . The present study was not designed to separate these two possible migrations, so all migration data refer to the movement of the implant in relation to the femoral bone.
It is recommended that the measurement precision in each particular study should be defined using double examinations . We calculated the precision levels by performing 22 double examinations at the 2-year follow-up. Each patient rose from the X-ray table and walked between the two sequential examinations within 10 min (Table 5).
The right hand side was used as the reference for the coordinate system when defining the micromotions . Thus, they rotation,z rotation, andx translation values for the left hips were changed to relate to a right-handed coordinate system (Fig. 1).
The patients returned for follow-up at 3, 6, 12, 24, and 60 months. The numbers of RSA and clinical examinations at different time intervals are presented in Table 3. At each follow-up, the data from standard radiographs, RSA, and Merle d’Aubigné-Postel were recorded. The Merle d’Aubigne-Postel score (which measures pain, mobility, and ability to walk) has a maximum possible score of 18.
The independent samples two-tailed t test was used to analyze the differences in migration between the two cement groups at 5 years postoperatively.Q–Q plots were used to verify the normality of the distribution. The measurement data from the two groups were subsequently pooled, and descriptive data were calculated for the whole patient population.
Here,x represents the differences between the double examinations, n = 22.
Finally, the precision was obtained by multiplying the SD by 2.074 (representing the 0.975 % point in at22 distribution).
Patient demographic data
At 5-year follow-up
Number of patients (hips)
Merle d’Aubigné-Postel score, mean (range)
Reasons for not attending (n)
Did not want to attend (4)
Missed the RSA postop measurement (4)
Condition number exceeded 130 (1)
Mean error of rigid body fitting and condition number
Mean error of rigid body fitting (ME)
Condition number (CN)
Number of RSAs and clinical examinations at different time intervals
Number of examinations
24 months (precision)
Examination of the 5-year follow-up conventional radiographs showed no osteolysis or radiological signs of loosening.
Migration data at the 5-year follow-up (mean values) along with corresponding p values for the two-tailed independent samples t test comparing the two cements
This follow-up study describes the migration pattern of the Charnley flanged 40 femoral stem. A total of 35 patients (36 hips) were enrolled, and data from 23 patients (23 hips) at 5 years were obtained. The patients were followed and evaluated by RSA as well as by clinical examination. The Merle d’Aubigné-Postel score at 5 years indicated that the patient group was functioning well. None of the patients lost to follow-up had their hips revised.
Originally the study was designed to compare stem stability upon using SmartSet HV and Palacos R bone cement in a prospective randomised 2-year follow-up study. No differences in stem stability were found . In the present study at 5-year follow-up, both cements also performed equally well. The most striking finding at this point was the observation of the continuous migration of a composite construction. Stem subsidence (translation along they-axis) and stem retroversion (rotation around they-axis) continued progressively up to 5 years postoperatively. Consequently, the main focus was then to describe the overall stem migration at 5 years regardless of the cement used.
Precision of the RSA measurements
As seen in Table 5, the precision levels vary considerably between the 3 rotational migrations measured. The tantalum markers mounted on the implant define a polygon. Evaluating the scattering of the markers is vital in order to determine the migration of the implant (polygon) in relation to the rotational axes. The method is more sensitive to rotations when the markers are located far from the rotational axes. Good scattering of the markers improves the level of precision accordingly. This phenomenon is generally seen in RSA measurements of hip implants.
Our results show that most of the rotations and translations take place initially. The translation along and the rotation around they-axis represent stem subsidence and retroversion, respectively (Fig. 2b, e). The resultant joint force that is transferred by the prosthetic head and acts on the stem will tend to push the stem distally and rotate the stem internally inside the cement mantle. Consequently, the migration pattern presented herein should be expected. The results presented in Fig. 2b, e indicate persistent stem migration for both subsidence and internal rotation. Even though the subsidence is minor, it indicates persistent migration. The rotation into varus also increases, and was 0.18° at 5 years postoperatively (Fig. 2f).
All 6 migrations at 5 years in the present study are equal to or smaller than those noted in the 2-year follow-up Charnley study by Hallan et al.  (a longer follow-up period is not reported). Our 5-year data confirm this inclination.
We found the 5-year translation values to be considerably lower than what Onsten et al. described in a 2-year follow-up of 65 Charnley stems, where most of the migration took place during the first 3 months. Mean migration along they-axis was 0.24 mm, along thex-axis 0.31 mm, and along thez-axis 0.67 mm. The rotations were not reported .
Different stem designs are expected to migrate differently [8, 15]. Kärrholm et al.  believe that the quantity of subsidence after 2 years is the best predictor of medium- to long-term loosening. Subsidence of the femoral head of 1–2 mm during the first 2 postoperative years indicated an increased risk of loosening and revision. The postoperative examinations were performed over a time interval stretching from the first day to 2 months after surgery. The patient population consisted of both primary THR and revisions. The implant was the cemented Lubinus SP I prosthesis . Several factors—type of implant, follow-up time, time for baseline RSA, and precision level—mean that Kärrholm et al.’s study and results are not directly comparable to those from our study.
Ryd et al. argues that a migration exceeding 0.2 mm at 2 years implies an increased risk of implant loosening. This study on knees evaluated several brands of (cemented and uncemented) knee prostheses. Mechanical loosening occurred exclusively in prostheses which migrated continuously . Our study is not straightforward comparable to their study, since there are fundamental differences between hip and knee replacement, and Ryd et al. used maximum total point motion (MTPM) to identify migration, while we used orthogonal translations and rotations to describe segment motions.
The study of Ryd et al. dealt with knee prostheses, while that of Kärrholm et al. evaluated primary and revision hips. This may explain the large disparity between their two different limits for increased risk of loosening, which differ by factors of 10, 1–2 mm, and 0.2 mm in the two studies, respectively [6, 7]. This exemplifies the importance of more implant-specific migration data that can be used to predict long-term implant survival based on short-term migration data.
The Exeter stem is constructed as a loaded, double-tapered, collarless, polished stem, and is designed to subside, in contrast to the Charnley stem . A study with the Exeter stem found a mean subsidence of 1.25 mm at the 2-year follow-up. The mean internal rotation was 0.6° and the varus alignment was 0.21° . Another follow-up study of the Exeter stem showed continuing subsidence (1.77 mm, median) and retroversion (1.6°, median) up to 5 years. This may be expected with this type of fixation .
The Charnley stem is not supposed to migrate but to show immediate stability due to its composite construction . In spite of this, studies have confirmed that the Charnley stem does migrate and rotate [13, 16, 20]. Our study supports these findings and also demonstrates that migration continues until at least 5 years.
A number of earlier RSA studies on hips differ from current guidelines in the standardization of RSA investigation. The different measuring methods, follow-up intervals, and type of implants used constitute an obstacle to any comparison of these heterogeneous results. Planned RSA studies ought to act in accordance with published guidelines for the standardization of RSA  so that it is easier to compare them.
Since several factors appear to influence the level of migration, it may be difficult to define acceptable limits on implant short-term migration with respect to long-term survival . More implant-specific RSA background data are needed to predict what can be considered to be acceptable micromotion. Migration data are continuous.
In conclusion, we have found that the Charnley stem continued to subside and to rotate in retroversion up to 5 years after surgery in a well-functioning patient group. More implant-specific RSA data are needed to predict long-time implant survival based on short-term observations. The migration data presented herein could contribute to this, serving as reference data for a normal migration pattern of a Charnley flanged 40 prosthesis.
Conflict of interest
This article is distributed under the terms of the Creative Commons Attribution License which permits any use, distribution, and reproduction in any medium, provided the original author(s) and the source are credited.
- Garellick G, Kärrholm J, Rogmark C, Herberts P (2009) Swedish Hip Arthroplasty Register; annual report 2009. Swedish Hip Arthroplasty Register, Gothenburg
- Davidson D, de Steiger R, Graves S, Tomkins A (2010) National Joint Replacement Register; annual report 2010. Australian Orthopaedic Association, Adelaide
- Engesæter L, Furnes O, Havelin L, Fenstard A (2010) The Norwegian Arthroplasty Register. Department of Orthopaedic Surgery, Haukeland University Hospital, Bergen
- Karrholm J, Herberts P, Hultmark P, Malchau H, Nivbrant B, Thanner J (1997) Radiostereometry of hip prostheses. Review of methodology and clinical results. Clin Orthop Relat Res (344):94–110
- Malchau H (2000) Introducing new technology: a stepwise algorithm. Spine (Phila Pa 1976) 25:285View ArticleGoogle Scholar
- Karrholm J, Borssen B, Lowenhielm G, Snorrason F (1994) Does early micromotion of femoral stem prostheses matter? 4–7-year stereoradiographic follow-up of 84 cemented prostheses. J Bone Joint Surg Br 76:912–917PubMedGoogle Scholar
- Ryd L, Albrektsson BE, Carlsson L, Dansgard F, Herberts P, Lindstrand A, Regner L, Toksvig-Larsen S (1995) Roentgen stereophotogrammetric analysis as a predictor of mechanical loosening of knee prostheses. J Bone Joint Surg Br 77:377–383PubMedGoogle Scholar
- Shen G (1998) Femoral stem fixation. An engineering interpretation of the long-term outcome of Charnley and Exeter stems. J Bone Joint Surg Br 80:754–756PubMedView ArticleGoogle Scholar
- Mirza SB, Dunlop DG, Panesar SS, Naqvi SG, Gangoo S, Salih S (2010) Basic science considerations in primary total hip replacement arthroplasty. Open Orthop J 4:169–180. doi:10.2174/1874325001004010169PubMedPubMed CentralView ArticleGoogle Scholar
- Derbyshire B, Prescott RJ, Porter ML (2009) Notes on the use and interpretation of radiostereometric analysis. Acta Orthop 80:124–130. doi:10.1080/17453670902807474PubMedPubMed CentralView ArticleGoogle Scholar
- Kobayashi A, Donnelly WJ, Scott G, Freeman MA (1997) Early radiological observations may predict the long-term survival of femoral hip prostheses. J Bone Joint Surg Br 79:583–589PubMedView ArticleGoogle Scholar
- Malik MH, Fisher N, Gray J, Wroblewski BM, Kay PR (2005) Prediction of Charnley femoral stem aseptic loosening by early post-operative radiological features. Int Orthop 29:268–271. doi:10.1007/s00264-005-0667-1PubMedPubMed CentralView ArticleGoogle Scholar
- Husby OS, Haugan K, Benum P, Foss OA (2010) A prospective randomised radiostereometric analysis trial of SmartSet HV and Palacos R bone cements in primary total hip arthroplasty. J Orthop Traumatol 11:29–35. doi:10.1007/s10195-010-0087-xPubMedPubMed CentralView ArticleGoogle Scholar
- Selvik G (1989) Roentgen stereophotogrammetry. A method for the study of the kinematics of the skeletal system. Acta Orthop Scand Suppl 232:1–51PubMedView ArticleGoogle Scholar
- Huiskes R, Verdonschot N, Nivbrant B (1998) Migration, stem shape, and surface finish in cemented total hip arthroplasty. Clin Orthop Relat Res (355):103–112
- Hallan G, Aamodt A, Furnes O, Skredderstuen A, Haugan K, Havelin LI (2006) Palamed G compared with Palacos R with gentamicin in Charnley total hip replacement. A randomised, radiostereometric study of 60 hips. J Bone Joint Surg Br 88:1143–1148. doi:10.1302/0301-620X.88B9.18008Google Scholar
- Stefansdottir A, Franzen H, Johnsson R, Ornstein E, Sundberg M (2004) Movement pattern of the Exeter femoral stem; a radiostereometric analysis of 22 primary hip arthroplasties followed for 5 years. Acta Orthop Scand 75:408–414. doi:10.1080/00016470410001169-1PubMedView ArticleGoogle Scholar
- Valstar ER, Gill R, Ryd L, Flivik G, Borlin N, Karrholm J (2005) Guidelines for standardization of radiostereometry (RSA) of implants. Acta Orthop 76:563–572. doi:10.1080/17453670510041574PubMedView ArticleGoogle Scholar
- Biau DJ, Brand RA (2009) Robert Merle d’Aubigne, 1900–1989. Clin Orthop Relat Res 467:2–6. doi:10.1007/s11999-008-0571-2PubMedPubMed CentralView ArticleGoogle Scholar
- Onsten I, Akesson K, Besjakov J, Obrant KJ (1995) Migration of the Charnley stem in rheumatoid arthritis and osteoarthritis. A Roentgen stereophotogrammetric study. J Bone Joint Surg Br 77:18–22Google Scholar
- McCalden RW, Charron KD, Yuan X, Bourne RB, Naudie DD, MacDonald SJ (2010) Randomised controlled trial comparing early migration of two collarless polished cemented stems using radiostereometric analysis. J Bone Joint Surg Br 92:935–940. doi:10.1302/0301-620X.92B7.24462PubMedView ArticleGoogle Scholar