Perspectives on metaphyseal conservative stems
© Springer-Verlag Italia 2008
Received: 6 April 2007
Accepted: 8 October 2007
Published: 13 March 2008
Total hip replacement is showing, during the last decades, a progressive evolution toward principles of reduced bone and soft tissue aggression. These principles have become the basis of a new philosophy, tissue sparing surgery. Regarding hip implants, new conservative components have been proposed and developed as an alternative to conventional stems. Technical and biomechanical characteristics of metaphyseal bone-stock-preserving stems are analyzed on the basis of the available literature and our personal experience. Mayo, Nanos and Metha stems represent, under certain aspects, a design evolution starting from shared concepts: reduced femoral violation, non-anatomic geometry, proximal calcar loading and lateral alignment. However, consistent differences are level of neck preservation, cross-sectional geometry and surface finishing. The Mayo component is the most time-tested component and, in our hands, it showed an excellent survivorship at the mid-term follow-up, with an extremely reduced incidence of aseptic loosening (partially reduced by the association with last generation acetabular couplings). For 160 implants followed for a mean of 4.7 years, survivorship was 97.5% with 4 failed implants: one fracture with unstable stem, 1 septic loosening and 2 aseptic mobilizations. DEXA analysis, performed on 15 cases, showed a good calcar loading and stimulation, but there was significant lateral load transfer to R3–R4 zones, giving to the distal part of the stem a function not simply limited to alignment. Metaphyseal conservative stems demonstrated a wide applicability with an essential surgical technique. Moreover, they offer the options of a “conservative revision” with a conventional primary component in case of failure and a “conservative revision” for failed resurfacing implants.
KeywordsArthroplasty Conservative Hip Metaphyseal DEXA
Total hip replacement is one of the most popular surgical procedures, with a worldwide successful application related to its high standardized outcome. Nevertheless, the clinical and functional results of prostheses have shown, during the last decades, a significant improvement that has gradually pushed their performances beyond their previous limits.
This continuing evolution has progressively modified the morphology, materials, surface finishing and tribologic coupling of the implants, while the widening of indications to increasingly younger patients has led clinicians to experiment with conservative options that preserve as much bone stock as possible.
This last point has become part of a new philosophy, inspired by principles of conservation (both related to surgical approaches and prosthetic implants), called Tissue Sparing Surgery . In particular, bone retaining is based upon the question: why sacrifice an intact anatomic structure, if it can be useful to implant stability [2, 3], joint biomechanics or a future revision procedure? Besides, this statement, giving an implicit contraindication in maintaining altered anatomic structures (both congenital and acquired), suggests by itself what could be a proper use of conservative options.
Rationale and design of metaphyseal stems
Excluding resurfacing implants, which cannot be considered stems due to their particular characteristics, there are two major groups of conservative implants: neck-retaining and metaphyseal bone-stock-preserving stems. The last group, object of this paper, was developed after the understanding of mechanical properties and role of cortical neck and metaphyseal cancellous bone in stress transfer and implant stabilization (primary and biologic fixation) .
In 1993, Jasty et al.  noted, in an experimental study, that the diaphyseal tract of a traditional femoral component was no longer necessary to implant stability once a proximal fixation was achieved. A few years later, Whiteside et al.  tested in vitro different levels of femoral neck preservation and their ability to neutralize torsional forces of femoral components: preservation of 50% or more of cortical neck was able to produce acceptable micromotions, while 15% or less produced macromotion with conventional designs.
All these biomechanical considerations led to the birth, in 1984, of a new femoral solution, the Mayo stem , (Zimmer, Warsaw, USA), which has been modified over the following years to the actual component. This prosthetic implant is described as an ultrashort, straight and multitapered component that, according to its developer B.F. Morrey, does not need to look like the femur . The Mayo stem has a proximal wide trapezoidal cross-section functional to search a multipoint contact within cortical femoral bone. This innovative geometry is able to stabilize the implant and to preserve proximal cancellous bone (impacted within a cortical shell, to increase torsional and axial stability). For these reasons, an appropriate level of neck osteotomy, able to leave an intact cortical ring in which cancellous bone could be impacted, is mandatory (Fig. 1).
The Mayo stem represents an evolution of principle, already time-tested for conventional straight stems (and recently confirmed by Swanson ) of “fit without fill”, with necessary modification of proximal shape and fixationto achieve torsional and axial stability. In fact, the extremely reduced diaphyseal segment of this implant is functional just to varus-valgus alignment along the lateral femoral cortex, leaving the diaphyseal canal practically virgin (Fig. 2).
These apparently limited variations involve, nevertheless, some differences in technical execution and fixation principles. The Nanos stem requires a higher neck osteotomy and more proximal filling to achieve, with a rounded cross-section, a primary stability, both axial and torsional: this last aspect apparently reduces the validity of proximal cancellous bone preservation (Fig. 4). The Metha stem presents an overall reduction of dimensions (length, cross-section area) compared to the Mayo stem, increasing the amount of retained bone but, on the other hand, requiring an extremely accurate reproduction of neck osteotomy to a level able to allow a proper spongy bone impaction and torsional stability.
Both these implants offer, doubtless, an evolution in surface finishing: a circumferential, last generation osteoinductive coating replaces the grit-blasting with mesh pads offered by the Mayo stem (Fig. 5), criticized for a potential risk in creating sleeves for distal debris migration (on our advice, initially developed to ensure the maximum mechanical stability at the corner of the trapezoidal cross-section, leaving to the mesh pads the role of secondary bone integration). However, an analysis of a single, explanted, radiographically stable Mayo stem did not detect significant differences in bone integration rate, compared to a conventional cementless implant .
Our experience started in 1999 with the Mayo stem (160 implants, with follow-up ranging from 1 to 7 years) and recently extended to Nanos (2005–2006, 25 implants) and Metha (2006, 27 implants).
Analysis of survivorship, clinical results and complications can be performed only for the most time-tested Mayo stem, but we will also share our preliminary data for the Nanos and Metha stems.
The survivorship of the 160 Mayo stems after a mean of 4.7 years (range, 1–7 years), was 97.5%. Mean age of patients at the implantation was 63.4 years (range, 42–83 years). Main diagnosis was primary arthritis (129 cases, 80.6%), osteonecrosis (15 cases, 9.4%), post-traumatic arthritis (9 cases, 5.6%), Crowe type I congenital hip dysplasia (CHD; 3 cases, 1.9%), failed osteosynthesis of proximal femur (2 cases, 1.3%) and failed resurfacing arthroplasty (2 cases, 1.3%).
Four implants failed. There was 1 early postoperative fracture (6 weeks postoperatively) with unstable stem (B2 in Vancouver classification), 1 septic loosening, and 2 aseptic mobilizations after 4 and 5 years. All these 4 implants have been revised. The fracture required a distaly stabilized stem (Wagner stem, Zimmer), the septic loosening received a two-step revision (antibiotic spacer followed by a primary Versys stem, Zimmer) and the two aseptic mobilizations were revised directly with a conventional primary implant (Platform, Smith&Newphew, Memphis, USA).
Two more stems are actually closely watched over because of an intra-operative perforation at the tip of the implant, but they show to be clinically and radiographically stable after 3 and 5 years. Other complications detected were 3 intra-operative proximal femoral cracks and 1 intra-operative diaphyseal crack (the last related mainly to a minimally invasive Watson-Jones procedure, with improper femoral manipulation); all were treated with metallic cerclage and not further operated.
One case presented with superolateral thigh pain (a complication usually not detected with this implant ) that was related to a cortical stem tip impingment (seen radiographically and with Tc99 scintigraphy). This resolved with pulsed electromagnetic field (PEMF) treatment for 1 month.
Periprosthetic bone mineral density (BMD) data in 7 regions of interest. Values are mean (SD) (Modified from )
Region of interest
This result agrees with the radiological appearances of some cases of lateral cortex hipertrophy (however not constant and apparently not related to surgical variables such as under-sizing of the component in the proximal femur) and rare cases of trochanteric stress-shielding (R1–R2) (Fig. 6). A widening of the study to a larger sample and a further clinical and radiographic investigation may produce more definitive results on the biotrophism of this femoral component. About this point, another comparative study  found that the shortest component tested (Mayo) out of four implant design showed the lowest changes in bone density (however they were again summarized as calcar resorption and distal hipertrophy).
Encouraging results obtained at short- to mid-term follow-up, such as low failure rate (2.5%) and a low specific complication rate (4.3%), and the wide versatility to different joint reconstruction indications (practically only mild to severe proximal changes, as Crowe type II CHD or more, severe post-surgical modifications such as femoral osteotomy, senile or post-menopausal severe osteoporosis were excluded) in middle-aged patients (range, 55–75 years) make the Mayo stem one of our favorite options in elective adult reconstruction (regularly coupled in the last 3 years with a large diameter femoral head), with a margin of applicability even in intracapsular fractures with at least 15% of the femoral neck intact (18 cases operated, 11.3% of total).
More recently in our experience, the Metha stem seems to offer the same clinical and technical advantages with the addition of femoral neck modularity. Even if majority of the cases can be managed with a standard neck (135° neck-shaft angle, 0° anteretro-version) in particular situations a modular cervical component may avoid a negative change in bio-mechanic static parameters. Four cases (14.8%) inour experience with Metha stems were successfully treated with a non-standard cervical modularity (1 case with 7.5° of retro-version, 2 hip replaced with 140° and 1 hip with 130° of neck-shaft angle). This option may extend the indication to mild proximal femur changes and, retrospectively, would have avoided 4 cases of lateral off-set reduction detected with a Mayo stem (provided a fixed neck-shaft angle). The need for a higher osteotomy, as discussed fractures (only 2 cases in our preliminary experience, 7.4% of total).
Metaphyseal conservative stems provide undoubled advantages in term of revision. As directly experienced with the Mayo stem, they allow a “conservative revision” in case of failure, with conversion to traditional primary implant (Fig. 7). Moreover, they can represent a conservative option for more conservative failed implants such as resurfacing (revised in our experience with the Mayo stem ) (Fig. 8), or CFP (1 case revised in our experience with a Nanos stem (Fig. 9).
Metaphyseal preserving implants present a biomechanic rationale, a good clinical mid-term behavior and a perspective of easy revision able to place them in the foreground of contemporary hip replacement, even if it is hard to affirm that they may substitute conventional implants or preserving implants with excellent long-term follow-up (such as CFP ).
On our advice, they should be elected as the operative solution after more conservative implants have been excluded and a careful balance of epidemiologic (senile and post-menopausal cortical and trabecular bone loss curves) and bio-morphologic parameters (flare index, DEXA analysis) has been performed. A mis-understanding of these parameters may lead, as we unfortunately experienced with both metaphyseal and neck-retaining implants, to rare but dramatic early failures.
- Pipino F, Keller A (2006) Tissue-sparing surgery: 25 years’ experience with femoral neck preserving hip arthroplasty. J Orthop Traumatol 7(1):36–41View ArticleGoogle Scholar
- Carlson L, Albrektsson B, Freeman MA (1988) Femoral neck retention in hip arthroplasty. A cadaver study of mechanical effects. Acta Orthop Scand 59(1):6–8PubMedView ArticleGoogle Scholar
- Pipino F, Calderale PM (1987) Biodynamic total hip prosthesis. Ital J Orthop Traumatol 13(3):289–297PubMedGoogle Scholar
- Jasty M, Krushell R, Zalenski E, O’Connor D, Sedlacek R, Harris W (1993) The contribution of the nonporous distal stem to the stability of proximally porous-coated canine femoral components. J Arthroplasty 8(1):33–41PubMedView ArticleGoogle Scholar
- Whiteside LA, White SE, McCarthy DS (1995) Effect of neck resection on torsional stability of cementless total hip replacement. Am J Orthop (10):766–770Google Scholar
- Morrey BF (1989) Short-stemmed uncemented femoral component for primary hip arthroplasty. Clin Orthop Relat Res (249):169–175Google Scholar
- Morrey BF, Adams RA, Kessler M (2000) A conservative femoral replacement for total hip arthroplasty. A prospective study. J Bone Joint Surg Br 82(7):952–958PubMedView ArticleGoogle Scholar
- Swanson TV (2005) The tapered press fit total hip arthroplasty: a European alternative. J Arthroplasty 20[4 Suppl 2]:63–67PubMedView ArticleGoogle Scholar
- Meldrum RD, Willie BM, Bloebaum RD (2003) An assessment of the biological fixation of a retrieved Mayo femoral component. Iowa Orthop J 23:103–107PubMedPubMed CentralGoogle Scholar
- Albanese CV, Rendine M, De Palama F et al (2006) Bone remodelling in YHA: a comparative DXA scan study between conventional implants and a new stemless femoral component. A preliminary report. Hip Int 16:9–15PubMedGoogle Scholar
- Roth A, Richartz G, Sander K et al (2005). Periprosthetic bone loss after total hip endoprosthesis. Dependence on the type of prosthesis and properative bone configuration. Orthopade 34(4):334–344PubMedView ArticleGoogle Scholar
- Berger RA (2004) The technique of minimally invasive total hip arthroplasty using the two-incision approach. Instr Course Lect 53:149–155PubMedGoogle Scholar
- Bertin KC, Rottiger H (2004) Anterolateral mini-incision hip replacement surgery: a modified Watson-Jones approach. Clin Orthop Relat Res (429):248–255Google Scholar